Gas Chromatography - Gas Supplies > Pressure Controllers > Page 7
The first control on any gas line is afforded by a simple pressure controller. There are a number of pressure controllers associated with a gas chromatograph. The reducing valves on the gas tanks are examples of simple pressure controllers and the flow controllers that are used for detector and column flow control often involve devices based on the same principles. A diagram of a pressure controller is shown in figure 2.
Figure 2 The Pressure Controller
The pressure controller consists essentially of two chambers separated by a diaphragm, in the center of which is a needle valve that is actuated by the diaphragm. The diaphragm is held down by a spring that is adjustable so that the pressure in the second chamber, and thus the outlet flow, can be set at any chosen value. When gas enters the lower chamber, the pressure on the lower part of the diaphragm acts against the spring setting, and opens the valve. Gas then passes into the upper chamber and pressure is built up in the upper chamber to the value that has been set at which time the diaphragm moves downward closing the valve. If the pressure falls in the upper cylinder, the diaphragm again moves upward due to the pressure in the lower chamber, which opens the valve and the pressure in the upper chamber is brought back to its set value.